topicarray["td20e89"] = {"Html":"\u003c?xml version=\"1.0\" encoding=\"UTF-16\"?\u003e\u003chtml xmlns:msxsl=\"urn:schemas-microsoft-com:xslt\"\u003e\u003cdiv class=\"main_body\"\u003e\u003cbr/\u003e\u003cdiv class=\"content_body\"\u003e\u003cdiv class=\"body_headline\" style=\"display:none;\"\u003eBağla\u003c/div\u003e\u003cdiv class=\"prolog-container\"/\u003e\u003cdiv class=\"expandable-container\" data-type=\"condition\"\u003e\u003cdiv class=\"h4 block-title header-object\" data-expandable=\"minus\"\u003e\u003cspan class=\"data-expandable\"\u003eBağlantı tekniği\u003c/span\u003e\u003c/div\u003e\u003cdiv class=\"expandable-content\" style=\"display: block;\"\u003e\u003cdiv class=\"p p-default floating-text\"\u003e\u003cspan\u003eElektriğin ve sinyal çıkışının bağlantısı gövdedeki yay baskılı klemenslerle yapılır.\u003c/span\u003e\u003c/div\u003e\u003cdiv class=\"safetycontainer inserted-object\"\u003e\u003ctable width=\"100%\" class=\"note\"\u003e\u003ctbody\u003e\u003ctr valign=\"top\"\u003e\u003ctd colspan=\"3\" class=\"hint-headerContent\"\u003e\u003cp\u003e\u003cdiv style=\"display:table;\"\u003e\u003cdiv style=\"display:table-cell;vertical-align:middle;\"\u003eNot\u003c/div\u003e\u003c/div\u003e\u003c/p\u003e\u003c/td\u003e\u003c/tr\u003e\u003ctr class=\"hint-content\"\u003e\u003ctd class=\"emptyIcon\"/\u003e\u003ctd class=\"emptyIcon\"/\u003e\u003ctd class=\"hint-content-text\"\u003e\u003cdiv\u003e\u003cdiv class=\"p p-default floating-text\"\u003e\u003cspan\u003eGerek sabit kablolar gerekse telli kılıflar içindeki esnek kablolar klemens ağzına bağlanabilir. Esnek kablolarda klemenslerin açılabilmesi için bir tornavida (3 mm kesici eni) ile tetikleme kolunu klemens ağzından dışarı yöne itin. Kolu serbest bıraktığınızda klemensler yeniden kapanır.\u003c/span\u003e\u003c/div\u003e\u003c/div\u003e\u003c/td\u003e\u003c/tr\u003e\u003ctr\u003e\u003ctd colspan=\"3\" class=\"hint-lower-line\"/\u003e\u003c/tr\u003e\u003c/tbody\u003e\u003c/table\u003e\u003c/div\u003e\u003cdiv class=\"safetycontainer inserted-object\"\u003e\u003ctable width=\"100%\" class=\"note\"\u003e\u003ctbody\u003e\u003ctr valign=\"top\"\u003e\u003ctd colspan=\"3\" class=\"hint-headerContent\"\u003e\u003cp\u003e\u003cdiv style=\"display:table;\"\u003e\u003cdiv style=\"display:table-cell;vertical-align:middle;\"\u003eNot\u003c/div\u003e\u003c/div\u003e\u003c/p\u003e\u003c/td\u003e\u003c/tr\u003e\u003ctr class=\"hint-content\"\u003e\u003ctd class=\"emptyIcon\"/\u003e\u003ctd class=\"emptyIcon\"/\u003e\u003ctd class=\"hint-content-text\"\u003e\u003cdiv\u003e\u003cdiv class=\"p p-default floating-text\"\u003e\u003cspan\u003eTellerin yalıtımlarının en az 0,5 mm veya daha üstünde bir kuvvete sahip olması gerekir.\u003c/span\u003e\u003c/div\u003e\u003c/div\u003e\u003c/td\u003e\u003c/tr\u003e\u003ctr\u003e\u003ctd colspan=\"3\" class=\"hint-lower-line\"/\u003e\u003c/tr\u003e\u003c/tbody\u003e\u003c/table\u003e\u003c/div\u003e\u003cspan class=\"img_free inserted-object has-image-title\"\u003e\u003cdiv style=\"margin-bottom:-4px;line-height:12.5pt;\"/\u003e\u003cdiv style=\"margin-bottom:-4px;line-height:12.5pt;\"/\u003e\u003cimg src=\"images/EA-Anschlussraum-Litzenanschluss-Zweileiter-4-20mA-PULSx1.svg\" docVersionId=\"contents\" language=\"tr\" class=\"img_free\" id=\"contents_td20e89_EA-Anschlussraum-Litzenanschluss-Zweileiter-4-20mA-PULSx1\"/\u003e\u003cdiv class=\"p imageTitle\"\u003e\u003cspan class=\"imageTitle\"\u003eBağlantı\u003c/span\u003e\u003c/div\u003e\u003c/span\u003e\u003cdiv class=\"p p-default floating-text\"\u003e\u003cspan\u003eMaks. tel kesiti ile ilgili daha fazla bilgi için \"\"\u003cspan class=\"i\"\u003eTeknik özellik - Elektromekanik bilgiler \u003c/span\u003e\" bölümüne bakın.\u003c/span\u003e\u003c/div\u003e\u003c/div\u003e\u003c/div\u003e\u003c/div\u003e\u003c/div\u003e\u003c/html\u003e"}